- Bring the pouched test device to room temperature (15-30℃) prior to testing. Do not open the pouch until ready to perform the assay.
- Remove the test device from the sealed pouch. Lay it on a flat, clean and dry surface.
- For serum or plasma specimens, hold the dropper vertically, transfer 2 drops of serum or plasma (approximately 50 µL) to the sample well (S) of the test device and start the timer.
- Read the result between 5-10 minutes. Do not interpret results after 20 minutes.
Two pink-colored lines appear in the result window, one in the control region "C" and another one in the test region "T". This indicates a positive result and the AFP having been detected.
Only the control line appears in the result window. This indicates a negative result and no AFP having been detected.
If no line appears in the control region, the test result is invalid regardless of the presence or absence of the test line.
Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for the control line failure. Review the procedure and repeat the test with a new device. If problem persists, please contact your local distributor.
|Description||Specimen||Format||Catalog No.||Sensitivity||Package Size||Strip Width||Certificate|
|(AFP) Alpha-Fetoprotein Test||Serum/Plasma||Strip||E10-11||20ng/mL||50 Tests||3.0 mm||CE|
|Cassette||E10-20||25 Tests||4.0 mm|
|Whole Blood||E10-21||25 Tests||4.0 mm|