- Bring the pouched test device to room temperature(15-30℃) prior to testing. Do not open the pouch until ready to perform the assay.
- Remove the test device from the sealed pouch. Lay it on a flat, clean and dry surface.
- Use the pipette packed in the pouch to draw and slowly add 1 drop of whole blood specimen to the sample well. Hold the buffer bottle vertically and add 1 drop to the sample well. / If using a pipette, change a new one to avoid cross-contamination. Draw and transfer 2 drops of buffer to the sample well.
- Wait for 10-15 minutes and read results. Do not read results after 20 minutes. Caution: The above interpreting time is based on room temperature range of 15-30℃. If your room temperature is significantly lower than 15℃, then the interpreting time should be properly increased to 30 minutes.
Positive:Two colored lines should be observed in the result window, one in the test region and another one in the control region.
Negative:Only the control line appears in the result window.
Invalid:No line appears in the control region regardless of the presence or absence of the line in the test region.
NOTE: Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for the control line failure. Review the procedure and repeat the test with a new device. If problem persists, please contact your local distributor.
|Description||Specimen||Format||Catalog No.||Sensitivity||Package Size||Strip Width||Certificate|
|(HCV) Hepatitis C Virus Test||Serum/Plasma||Strip||B20-11||99.53%||50 Tests||3.0 mm||CE2265|
|Cassette||B20-20||25 Tests||4.0 mm|
|Whole Blood||B20-21||25 Tests||4.0 mm||——|